Nexus 3000

SINGLE PARAMETER READING

The instrument's very large display allows all operating parameters to be viewed. The instrument can handle the following measurements: pH, Rx, Conductivity, Chlorine amperometric, mA input potentiostatic sensors , Flow
The following configurations are available:
- 144 x 144 mm Ip 65 for wall mounting
- 96 x 96 mm for panel mounting (with IP65 front and IP20 rear section)
It is possible to set the Relay2 output for the flushing function: through the Washing, Waiting for stabilisation and Interval of activation, the sensor cleaning process can be set to automatic mode.

GUIDED CALIBRATION

The calibration wizard allows you to follow all the steps and return the quality of the sensors for optimal alignment.

ON-OFF / TIMED / PROPORTIONAL RELAY ACTIVATION

The Relay output can be set in three different modes: On-Off activation, timed activation, proportional activation. These modes make it possible to find the right compromise in the management of chemical parameters within the process.

ANALOGUE AND DIGITAL OUTPUT

The Nexus 3000, is equipped with 2 relays for set point control and probe cleaning management, temperature measurement and alarm control, a 4/20ma analogue output and a frequency output. The installer can set the instrument by choosing the most suitable configuration for the system.

TEMPERATURE COMPENSATION

By using a temperature sensor, it is possible to display the temperature in the instrument's display and make the compensation in the intended measurements.

EVENTS AND ALARMS

Via the alarm list, it is possible to view the history of events recorded in the instrument. The entire list can be reset at any time.

The article Nexus 3000 comes from Injecta.